Aragor This can make isolating events of interest difficult. The farther you turn the outer control, the faster the zoom box moves. The bright, widescreen display shows a long time window. Waveforms are superimposed with one another and waveform points that occur more frequently are intensified. Key software applications — OpenChoice Desktop, and Microsoft Excel and Word toolbars — are included standard with each oscilloscope to enable fast and easy direct communication with your Windows PC.
|Published (Last):||23 July 2014|
|PDF File Size:||3.17 Mb|
|ePub File Size:||8.39 Mb|
|Price:||Free* [*Free Regsitration Required]|
The following three conditions must be met prior to performing these procedures: 1. Push the Self Test lower-bezel button. Wait while the self test runs. When the self test completes, a dialog box displays the results of the self test.
Push the Menu Off button to clear the dialog box and Self Test menu. Push the Source lower-bezel button. Select the AC Line trigger source with Multipurpose knob a. You do not need to connect an external signal to the oscilloscope for this DC Balance test. Push the front-panel Acquire button.
Push the front-panel Default Setup button to set the instrument to the factory default settings. Push the front-panel Horizontal Acquire button, then push the bottom-bezel Average button, and then push the side-bezel Average button, to turn averaging on.
Calculate V — V , and then enter V in the test record. Enter this measurement in the test record. Range 1. Connect the oscilloscope to a DC voltage source to run this test.
If using the Fluke calibrator as the DC voltage source, connect the calibrator head to the oscilloscope channel to test. Push the Set to 0V side-bezel button. Use the Vertical Position knob to place the trace at the bottom of the display -4 divisions.
Horizontal Position Time 1. If necessary, turn the Horizontal Position knob to set the delay to exactly 1. Compare the rising edge of the marker with the center horizontal graticule line. Push the side-bezel D7 - D0 button. Use Multipurpose knob a to set the D7-D0 threshold level to 0 V. Push the front-panel DD0 button.
This manual is also suitable for:.
Tektronix DPO2014 Manuals
Tektronix DPO2014B Manuals
Tektronix Diagrams, Schematics, Service Manuals